Recently Browsing 0 members
No registered users viewing this page.
See below for FLO's opinion of this camera.
Received new (from FLO) on 13 January, 2018. Used ONLY TWICE.
Reason for sale: The "cooled" version of this camera is being discontinued and FLO were offering their last one at a ridiculously low price, so I purchased that one (for easier OSC DSO imaging), making this camera surplus to requirements.
FLO price is £229 (as you can see) + £5.95 delivery, making a total of £234.95. I am offering this barely out of its box camera for £200 including delivery - that's more than 14% less than you would pay for a 7-week younger one. I have been advised that RM will not insure cameras, but it is still in its original packaging. Bank transfer only - sorry, NO paypal.
As stated, I have not used it much, but, for a "teaser" of what this camera may be capable, visit http://www.ne3filters.co.uk/g8kpd3NFcer4L/index.html - I will definitely be going back to this target with the cooled version!
PM me if interested.
90min HII, 72min [NII], and 80min [OIII] in 2 min subs with the 80mm f/4.4 and ASI1600 MMC camera through 3nm Astrodons. Piggy backed on the 130 f/7 mounted on the DDM60 and encoder guided.
Sigma Add stacking in AA5, then gradient removal, alignment and Trichromy with [NII] mapped to Red, HII to Green, and [OIII] to Blue, following the Hubble Palette.
Multiple Histogram Stretches, and a Low Pass filter to get rid of some of the noise. The RGB stack needed a serious crop as the [OIII] subs were at a steep angle to the other data, which is a pity as I can see plenty of outlying detail in the individual stacks.
I'm not taking this particular image any further as the data is too short. I've already got another 2 hours of HII in 5 min subs which I'll stack and add to the existing HII stack. The [OIII] I think I'll bin as it's so badly skewed, I'll try to replace it with 5 min subs properly aligned. The [NII] is OK as it stands, but I'll try to add to it.
When I'll be able to get more data is a moot point, as the weather here has been dire. There *may* be a couple of windows coming up, but I'm not counting on them.
I started this zwo 1600 problem topic yesterday, but I couldn't get the answer. I have found that my problem is temperature related. when the sensor chamber drops below zero the download hangs. Sorry for the new topic but i am getting really tired of this.
Could you please help me out?